5k Voltage Probe (set of 3)
Model No: 5KVPSet
Powersight’s 5KV voltage probe for direct connection to 600 to 5kv rms AC or DC.
Accuracy ±1% for voltages between 600 and 5,000 Vrms at 50Hz and 60Hz.
Probe length - 14 inches
Cable length - 5 feet
Weight - less than 1 lb.
Output declines less than 1 db from 50 to 900 Hz.
PowerSight Technology provides its model 5KVP voltage probes to allow complete analysis of 4,160 V power systems. These probes plug directly into PowerSight Technology’s PowerSight® Energy Analyzers.
PowerSight Technology introduces its new model 5KVP voltage probes capable of connecting directly to 4,160V. Traditionally, professionals who wished to measure 4,160V volt lines needed to locate a PT (potential transformer) and then connect to the low voltage output secondary of the PT.
The 5KVP allows direct connection to the primary voltage. Bypassing the PT avoids any phase shift and linearity problems introduced by it. It also allows direct connection at any point of interest whether PTs already exist or not. The probes are used exclusively with the PowerSight® line of Energy Analyzers, manufactured by PowerSight Technology Inc.
Each probe comes with a hook attachment to facilitate hands-free measurement and long-term monitoring of 5kV lines. Each probe has a ground clamp, a neutral connector and a reduced voltage connector. The connectors are safety banana plugs. Each lead is labeled to advise how to use it.
The 5KVP comes as a set of three probes. Each probe is 14 inches long and weighs less than a pound. The fully extended length of the leads and probes is 7 feet.
Using the 5K Voltage Probe
Whenever possible, connect the probes to a de-energized circuit.
First, connect the ground lead of
each of the probes to a secure ground point.
Next connect each of the Vn safety
banana plugs into the Vn input of PowerSight. These are
special stackable plugs, so all three plug into the same
input. Next plug The Va, Vb, and Vc safety banana plugs
into the Va, Vb, and Vc inputs of PowerSight.
At this point, press the “Calibra”
button on PowerSight PS3000 & PS4000 or "Input Ratios"
button on the PS3500 & PS4500 and follow the procedure
to enter an input ratio of 100:1 for Va, Vb, and Vc.
Lastly, connect the hooked tips of each probe to the correct phase and then re-energize the circuit.
Not the right voltage? Check out our other voltage probes.
You can use these high voltage probes with our meters: